The execution of the project was scheduled from April 22, 2009 to March 21, 2009; the purpose was to enhance the current communication structure of EPB, resolve linkage difficulties encountered in plants, coordinate the enhanced system functions at EPB, consult and assist plants in the transmission module renewal, verify the creditability of the monitoring data from CEMS, provide consultation and establish quality assurance operation, practice the quality assurance system and auditing operation, execute the Relative Accuracy Test Audit (RATA) and Calibration Gas Audit (CGA), compare the relativity between monitored and inspected data, and improve the efficiency of EPB system, quality assurance operation, and onsite inspection. The project also including flue, area surrounding, and fuel inspections of stationary pollution source. It to ensure that the emissions from stationary pollution source status of compliance standards. All works were executed according to the schedule during project operation, the overall project has greater than 94.0 %. Results and progresses were detailed in table 1.4-1.
During the project operation, besides periodical maintenance of EPB information system and assistance in document inspections, 22 onsite inspections for functions of the monitoring system were completed. The inspection-found defects in all plants were separately guided, improved and traced to ensure the full compliance of the operation of CEMS. Additionally, there were 8 RATA inspections performed, all the inspections results met the requirement standards, in addition to Hsinwu Mill of YFY ENTERPRISE CORP (P002) that the relative accuracy of SO2 over the standards of regulations. Our team then counseling for this plant improvement, after implement RATA on January 12, 2010 by itself. By the EPB pursuant to January 26, 2010 RATA report submitted to the approval of its improvement to complete. And also perform 8 CGA, all the results in line with the standard.
According to the statistical results of plants uploading their data, the effective monitoring percentage had all been greater than 90 % for each plant in every season. This was in compliance with the CEMS regulation (85 %). The uploading ratio also reached above 95 %, which achieved the real time uploading control effect. For the photo electronic industries, our team also provided plant regulation consultation services in addition to the regular document inspection. During project execution, we inspected and assisted applications from AUO, Changhua picture tube company, and Daxon Technology Inc..
The audits of stationary pollution sources have completed 30 stack inspections, 10 area surrounding inspections, 15 fuel analyses, and 861 examinations of analysis reports in internet. There are 9 sample over the standards of regulations and denounced by the EPB, requested to improve the operation of pollution in order to maintain good air quality.
The large display at the plaza in front of the Taoyuan County Government was in good condition under our maintenance works; besides the air quality and CEMS monitoring figures display, the screen also promote many government policies, totaling 180 broadcasts. People can get the information of guidance form (the Government by using) digital signage. And set up stationary pollution sources integrated system of forms of investigation, facilitate the EPB to immediately checking business cases.