The execution of the project was scheduled from April 22, 2007 to April 21,
2008; the purpose was to enhance the current communication structure of EPA,
resolve linkage difficulties encountered in public/private plants, coordinate the
enhanced system functions at EPA, consult and assist plants in the transmission
module renewal, verify the creditability of the monitoring data from continuous
auto monitoring equipments, provide consultation and establish quality
assurance operation, practice the quality assurance system and auditing
operation, execute the RATA inspection auditing, compare the relativity
between monitored and inspected data, and improve the efficiency of EPA
continuous auto monitoring system, quality assurance operation, and onsite
inspection. All work was executed according to schedule during project
operation; the results were as shown in table 1.4-1.
During the project operation, besides periodical maintenance of EPA
information system and assistance in document inspections, 21 onsite
inspections for functions of the monitoring system were completed. In addition,
all incompliance was traced and corrected under our assistance to ensure the full
compliance of the operation of the continuous auto monitoring system. Next,
there were 8 RATA inspections performed including Taoyuan Refinery of CPC
Corporation Taiwan (P002、P004、P005), Kuokuan Power Corporation P002,
Chungsan Power Corporation P003, Shinzon incineration plant P002, Jinhsin
plant, Nan Ya Plastic P501, and Huaya Cogeneration P101; all the inspections
results met the requirement standards.
In 2007, EPA had only conducted inspection on the CEMS linkage to the
CPC Taoyuan refinery and found no major flaws, which indicated that CEMS
control has been gradually achieving its goal under the supervision of EPA and
our inspection and consultation efforts.
For the photo electronic industries, this team also provided plant regulation
consultation services in addition to the regular document inspection. During project execution, we inspected and assisted applications from TPO display
company, AUO, Giantplus Technology, Changhua picture tube company, and
Hannstar. In the mean time, the CEMS training seminar had been held on Feb.,
19th, 2008; subject of volatile organic monitoring regulations had been
presented in the seminar, which would be helpful in promoting plant’s CEMS
During plants uploading their data, the effective monitoring percentage had
all been greater than 90% for each plant in every season. This was in
compliance with the CEMS regulation (85%). The uploading ratio also reached
above 95%, which achieved the real time uploading control effect.
The large display at the plaza in front of the city hall was in good condition
under our maintenance; besides the air quality and CEMS monitoring figures
display, the screen also promote many government policies, totaling 121
broadcasts, making the full use of the electronic bulletin board.